M3 RENISHAW, Styli straight, silicon nitride sphere, tungsten carbide shaft
Stylus straight M3, DK2 L16
Standard Styli suitable for all common measuring tasks. M3 thread for ZEISS RST, Renishaw SP and Hexagon HP-S sensors.
Not for ZEISS VAST XXT sensors (see product notes).
- SiN for less material deposits
- SiN spheres available from 0.3 up to 12 mm
- Tungsten carbide shaft with high rigidity
- Straight shaft with longer measuring length
- Proven ZEISS quality through high-end manufacturing
- Product Type
- Connection Type
- Ø Sphere (DK)
- 2.0 mm
- Length (L)
- 16.0 mm
- Ø Shaft (DS)
- 1.5 mm
- Ø Body (DG)
- 4.0 mm
- Measurement Length (ML)
- 10.0 mm
- Stylus Type
- Stylus Tip Geometry
- Stylus Tip Material
- Si. Nitride
- Shaft Material
- Tung. Carb.
- 0.8 g
It is not recommended to mix M3 RENISHAW elements with ZEISS XXT systems. The differnt base diameters may cause a loose connection, resulting in measuring errors.
Depending on the material of the stylus and the workpiece, wear or material deposits may collect on the stylus. Therefore, regularly check your styli under a microscope and clean or replace them if damaged. Depending on the type and degree of contamination, a wide range of cleaning agents are available. ZEISS application engineers recommend stylus cleaning wipes or microfiber cloths.